Payment & Shipping Terms:
|Height:||Max Height Of Specimen 175mm||Throat:||Max 165mm|
|Resolution:||0.1HR||Dwell Time:||0 - 60 Seconds, Adjustable|
|Printing:||Built-in Mini Printer||Hardness:||Conversion HV, HBW|
digital hardness tester,
hardness testing equipment
Digital Superficial Rockwell Hardness Tester with Built-in Printer and Hardness Conversion
iRock-435D Digital superficial Rockwell hardness tester equips with a newly-designed large displaying screen with good reliability, excellent operation and easy reading.
It can show and set the present scale, test force, test indenter, dwell time and hardness conversion.
The main functions as follows:
1, Support all the superficial Rockwell scales;
2, with hardness Conversion;
3, Test results can be saved and printed out,
4, with statistics functions like automatic calculation of maximum, minimum and average value;
5, With RS232 interface for data transferring to computer.
Suitable for surface quenched steel, surface heat treating and chemical treating materials, copper alloy, aluminum alloy, sheet, zinc layers, chrome layers, tin layers, bearing steel and cold and hard casting etc.
|Initial Test Force||3kgf (29.42N)|
|Total Test Force||15kgf (147.1N), 30kgf (294.2N), 45kgf (441.3N)|
|Indenter||Diamond Rockwell Indenter, ф1.588mmBall Indenter|
|Loading Method||Automatic (Loading/Dwell/Unloading)|
|Hardness Reading||LCD Screen Display|
|Test Scale||HR15N, HR30N, HR45N, HR15T, HR30T, HR45T, HR15W, HR30W, HR45W, HR15X, HR30X, HR45X, HR15Y, HR30Y, HR45Y|
|Conversion Scale||HV, HK, HRA, HRB, HRC, HRD, HRE, HRF, HRG, HRK, HR15N, HR30N, HR45N, HR15T, HR30T, HR45T, HS, HBW|
|Data Output||Built-in Printer,RS232 Interface|
|Max. Height of Specimen||175mm|
|Power Supply||AC220V/50Hz/1Ph, AC110V/60Hz/1Ph|
|Execute Standard||ISO 6508, ASTM E-18, JIS Z2245, GB/T 230.2|
|Dimension||520×215×700mm, Packing Dimension: 650×370×950mm|
|Weight||Net Weight: 73kg, Gross Weight: 100kg|
|Instrument Main Body||1 set||Diamond Rockwell Indenter||1 pc|
|ф1.588mmBall Indenter||1 pc||Middle Plane Test Table||1 pc|
|Large Plane Test Table||1 pc||V-shaped Test Table||1 pc|
|Hardness Block 70~85 HR30T||1 pc||Hardness Block 80~90 HR15N||1 pc|
|Hardness Block 65~80 HR30N||1 pc||Weight A, B, C||3 pcs|
|Power Cable||1 pc||Fuse 2A||2 pcs|
|Level||1 pc||Horizontal Regulating Screw||4 pcs|
|Screw Driver||1 pc||Spanner||1 pc|
|Anti-dust Cover||1 pc||Usage Instruction Manual||1 copy|
The Rockwell test consists of measuring the additional depth to which a carbide ball or diamond penetrator is forced by a heavy (major) load beyond the depth of a previously applied light (minor) load (SET point).
The minor load is applied first and a SET position is established on the dial gauge or displacement sensor of the Rockwell tester. Then the major load is applied. Without moving the piece being tested, the major load is removed and, with the minor load still applied, the Rockwell hardness number is automatically indicated on the dial gauge or digital display.
The diamond penetrator is used for testing materials such as hardened steels and cemented carbides. The carbide ball penetrators, available with 1/16 inch, 1/8 inch, 1/4 inch, and 1/2 inch diameter, are used when testing materials such as steel-copper alloys, aluminum and plastics to name a few.
Rockwell testing falls into two categories: Regular Rockwell testing (e.g., C and B scales) and Rockwell superficial testing (e.g., 30 N and 30 T scales).
High Rockwell hardness numbers represent hard materials and low numbers soft materials.
Contact Person: Mr. Andy DAI